You Need a Fume Hood…Now What?
Questions 1-8 on fume hoods
were in the July/August 2008 issue of Lab Manager Magazine.®
Here are twelve more things you should know...
9: What is a laboratory fume hood?
A laboratory fume hood is a ventilated enclosure where hazardous materials can be handled safely. The purpose of the fume hood is to contain contaminants and prevent their escape into the laboratory. This is accomplished by drawing contaminates within the hood’s work area away from the user, so that inhalation and contact are minimized. The fume hood is part of the laboratory fume hood system.
10: What is a laboratory fume hood system?
A laboratory fume hood system consists of four basic components: 1) base cabinet(s), 2) fume hood, 3) duct, and 4) remote blower. Each system is designed and installed for your specific ventilation requirements.
11: How do I know which fume hood I need?
There are two main factors to consider when selecting a fume hood style. First is the type of work being performed in the hood. Second is the air consumption of the hood. The types of fume hoods are airfoil by-pass, airfoil variable volume, airfoil add air, low flow, and other specialty fume hoods.
12: What is a by-pass fume hood?
The by-pass hood generally operates at a constant volume. As its sash is closed, the air entering the hood is re-distributed, therefore minimizing the high velocity air streams encountered in conventional hoods. The by-pass openings above and below the sash reduce fluctuations in face velocity as the sash is raised and lowered and the lower by-pass provides a continuous air sweep of the work surface.
13: What is variable volume hood?
The variable air volume (VAV) hoods are designed to be used with exhaust control systems provided by other manufacturers. In lieu of an upper by-pass, the VAV hood has a lintel which restricts the air intake of the hood to the sash opening and lower pass. The commercially available exhaust control system will detect the movement of the sash and adjust the volume of air required to maintain a constant face velocity.
14: How do I know what type of sash I need?
Most manufacturers offer the vertical rising sash with safety glass as a standard feature. However, horizontal sliding and combination horizontal/vertical sashes are also available. A combination horizontal/vertical sash can provide both safety and energy efficiency while the horizontal sliding panels act as a safety shield while reducing the sash opening by 50%. As with the selection of the proper hood, it is a good idea to consult with an expert in the fume hood industry to ensure the proper decision is made.
15: How do I know which fume hood liner my laboratory needs?
The most common liner available is made of epoxy resin. Other common liners include fiberglass, epoxy-coated steel, stainless steel, PVC, polypropylene and glass reinforced cement. Each liner material is suited for specific applications, and offers varying degrees of chemical resistance. Your specific application should be considered when making your selection.
16: What about fixtures?
Many people overlook or underestimate their fixture needs when ordering a fume hood. This is understandable given all the available options such as air, gas vacuum, water, electrical, and special needs fixtures. Look at the possibilities of need changes in the future also and consider the location of the fixtures to accommodate the requirement of a cup sink if one is necessary.
17: What type of duct do I need?
Duct, for the most part, may be the most confusing issue of your fume hood system. A number of types are available and choosing can be difficult from a list that includes poly-vinyl chloride, stainless steel, fiberglass, glass, and galvanized. The PVC remains the most popular given its high resistance to both acid and solvent vapors. Although PVDC duct is not inexpensive, its cost is considerably less than stainless steel, fiberglass, and glass. Extreme caution should be used if galvanized duct is considered.
18: Can’t my maintenance department install my fume hood?
There are many facility maintenance departments who are qualified to install a fume hood, but unfortunately, many fume hood systems are installed by unqualified installers. A fume hood must be properly installed and operated in order to achieve optimum performance. Qualified installation firms have been professionally trained in the many facets of a properly installed fume hood system. They understand why the duct run, blower location, stack height, damper system (if necessary), electrical, and plumbing connections are all vital links in your system’s performance. They know laboratory installations. Therefore, before you make the final decision on installing your new or retrofit system, contact your hood manufacturer to ensure the installation firm is authorized or certified with their products. Key questions should include:
How many systems has the installation firm installed?
Can they calibrate and certify the system’s performance?
Are they insured?
A final note: Do not assume that a firm familiar in HVAC systems is qualified in, or fully understands fume hood system installations.
19: After my system is installed, what’s next?
Before using your system, remember to create a chemical hygiene program
(CHP) as outlined by OSHA.2
Instruct your personnel on the proper operation of your fume hood and its upkeep. Your manufacturer and qualified installation firm can provide you with details. Training sessions are typically available and run for approximately one hour or less.
20: Why is preventative maintenance important?
Your fume hood systems should be evaluated periodically to ensure that they are functioning properly to ensure the safety of your personnel and your facility. Preventative maintenance on the fume hood motors and exhaust fans is not just something that you should do…. it is something that you must do! The average remote blower has over 20 moving parts that typically rotate over 136 million times a year and any number of these parts can weaken and eventually cause a premature system failure. Most failures occur without your knowledge, resulting in performance degradation and potentially place you and your operators at risk. Other items to be inspected and maintained are utilities, sashes and safety glass.
Of course there are many more than 20 questions surrounding fume hood systems, but hopefully this paper gives you guidelines that will be helpful with all aspects of choosing, installing and maintaining your systems. Fume hoods are designed to last you for many years, however if improperly used, even the best fume hood can be compromised beyond its capabilities. Key points to remember are, choose the proper hood for your application, rely on qualified firms for placement, installation and maintenance of your fume hood systems and comply with safety regulations.